Cardiac pacing serves to maintain or restore myocardial depolarization and thus ensure adequate cardiac output. In the ED, pacing corrects rhythm disturbances or starts in anticipation of a conduction problem with hemodynamic impact.1
Indications for emergency pacing are provided in Table 33-1.
TABLE 33-1Indications for Emergency Pacing |Favorite Table|Download (.pdf) TABLE 33-1 Indications for Emergency Pacing
|Indication ||Comments |
|Symptomatic or hemodynamically unstable bradycardia/AV block || |
Symptoms include hypotension, change in mental status, angina, and pulmonary edema.
Pharmacologic therapy may be used to temporize while preparing to pace.
|Severe sick sinus syndrome with prolonged asystole (generally >3 s) and syncope ||— |
|Ventricular standstill due to complete heart block or Mobitz type II AV block ||— |
|Torsades de pointes ||Overdrive pacing. |
|Recurrent monomorphic ventricular tachycardia || |
The technique is limited by:
Maximum pacing rate of the pacing device (usually 180 beats/min).
Potential of accelerating the ventricular tachycardia and inducing ventricular fibrillation.
|Unstable supraventricular tachycardia ||Overdrive pacing should be used only after pharmacologic intervention and cardioversion have failed. |
Cardiac pacemakers deliver an electrical stimulus to the heart through electrodes, causing depolarization and subsequent cardiac contraction.2 The modern implanted pacemaker only stimulates the heart chamber if it does not recognize (sense) intrinsic electrical activity from that chamber after a selected time interval. Impulses go to the atria, ventricles, or both.
Components of a cardiac pacemaker include:
Relevant clinical details of these components are provided in Table 33-2.
TABLE 33-2Pacemaker Component Details |Favorite Table|Download (.pdf) TABLE 33-2 Pacemaker Component Details
|Pacemaker Type ||Pulse Generator Location ||Electrode Location |
|Transcutaneous ||External || |
Skin of anterior chest wall and back or
Anterior chest wall below right clavicle and apex
|Transvenous ||External ||Venous catheter with tip in right ventricle and/or right atrium |
|Transesophageal ||External ||Esophagus |
|Epicardial ||External or Internal || |
Electrodes are usually placed on heart’s surface during surgery
|Permanent ||Internal (subcutaneous in the prepectoral region) ||Venous or epicardial |
Transcutaneous pacing is an emergency technique frequently chosen in patients presenting with hemodynamically significant bradycardia because of its easy application. It uses externally (chest wall) applied electrodes to deliver an electric impulse to stimulate the myocardium. Transcutaneous pacers differ from standard pulse generators: The pulse duration of the externally stimulating impulse is longer and the current output higher than in internal pacing. Muscle contraction (usually the chest wall or diaphragm) is notable during external pacing, especially at high outputs, and may be painful. The muscle twitching makes palpation of the radial, carotid, or femoral pulse difficult. Finally, ...